BOURDON GAUGE EPUB!
How does a Bourdon Gauge work? View our animation to gain a better understanding of the Bourdon Gauge. Other articles where Bourdon-tube gauge is discussed: pressure gauge: The Bourdon-tube gauge, invented about , is still one of the most widely used. Figure The Bourdon gauge. Bourdon Gauge The Bourdon gauge is shown in figure It works on the same principle as that of the snakelike, paper party.
|Published:||21 January 2015|
|PDF File Size:||15.84 Mb|
|ePub File Size:||15.31 Mb|
Bellows[ edit ] A pile of pressure capsules with corrugated diaphragms in an aneroid barograph In gauges intended to sense small pressures or pressure differences, or require that bourdon gauge absolute pressure be measured, the bourdon gauge train and needle may be driven by an enclosed and sealed bellows chamber, called an aneroid, which means "without liquid".
Early barometers used a column of liquid such bourdon gauge water or the liquid metal mercury suspended by a vacuum. This bellows configuration is used in aneroid barometers barometers with an indicating needle and dial cardaltimetersaltitude recording barographsand the altitude telemetry instruments used in weather balloon radiosondes.
These devices use the sealed chamber as a reference pressure and are driven by the external pressure. Other sensitive aircraft instruments such as air speed indicators and rate of climb indicators variometers have connections both to the internal part of the aneroid chamber and to bourdon gauge external enclosing chamber.
Bourdon tube pressure gauge
Magnetic coupling[ edit ] These gauges use the attraction of two magnets to translate differential pressure into motion of a dial pointer. As differential pressure increases, a magnet attached to either a piston or rubber diaphragm moves.
A rotary magnet that is attached to a pointer then moves in unison. To create different pressure ranges, the spring rate can be increased or decreased. Bourdon gauge gauge[ edit ] The spinning-rotor gauge works by measuring the amount a rotating ball is slowed by the viscosity of the gas being measured.
The ball is made of steel and is magnetically levitated inside a steel tube closed at one end and exposed to the gas to be measured at the other.
It is accurate and stable enough to be used as a secondary standard.
Bourdon-tube gauge | instrument |
The instrument requires some skill and knowledge to use correctly. Bourdon gauge corrections must be applied and the ball must be spun bourdon gauge a pressure well below the intended measurement pressure for five hours before using.
It is most useful in calibration and research laboratories where high accuracy bourdon gauge required and qualified bourdon gauge are available. Pressure sensor Piezoresistive strain gage Uses the piezoresistive effect of bonded or formed strain gauges to detect strain due to applied pressure.
Capacitive Uses a diaphragm and pressure cavity to create a variable capacitor to detect strain due to applied pressure. Magnetic Measures the displacement of a diaphragm by means of changes in inductance reluctanceLVDTHall effector by eddy current principle.
Piezoelectric Uses the piezoelectric effect in certain materials such as quartz to measure the strain upon the sensing mechanism due to pressure. Optical Uses the physical change of an optical fiber to detect strain due to applied pressure. Potentiometric Uses the motion of a wiper along a resistive mechanism to detect the strain caused by applied pressure.
Resonant Uses the changes in resonant frequency in a sensing mechanism to measure stress, or changes in gas density, caused by applied pressure. Thermal conductivity[ edit ] Generally, as a real gas increases in density -which may indicate an increase in pressure - its ability to conduct heat bourdon gauge.
In this type of gauge, a wire filament is heated by running current through it. A thermocouple or resistance thermometer RTD can then be used to measure the temperature of the filament. This temperature is dependent on the bourdon gauge at which the filament loses heat to the surrounding gas, and therefore on the thermal conductivity.
A common variant is the Pirani gaugewhich uses a single platinum filament as both the heated element and RTD. Pirani one wire [ edit ] Main article: Pirani gauge A Pirani gauge consist of a metal wire open to the pressure being measured.